Our smallest and most versatile eye tracker
The PCEye Mini can be used to fully replace the traditional keyboard and mouse with the Access package, which includes our software Windows Control for full computer access. If the ambition is to get started with eye tracking, the package Track & Learn is a good option. Choose product option below to see more information.
If you want to use the PCEye Mini with a tablet, please take a look at the EyeMobile Mini Bracket.
Product no: 111081
Experience greater personal freedom and independence by accessing all the functions of your Windows desktop or laptop computer using only your eyes.
PCEye Mini is a versatile eye tracker designed for all levels of eye gaze skills and needs. It can take you from paint, play and draw to full computer access.
Eye gaze access
Replace a traditional keyboard and mouse with your eyes and surf the web, connect on social media, play games, create documents and more.
If you're new to eye gaze, try a simple and efficient way to learn how to use it then measure and track your progress with Gaze Point and Gaze Viewer software.
Whether you’re just getting started with eye gaze or are ready to fully access your computer, we have a PCEye Mini package to meet your needs.
For full computer access, get PCEye Mini and Windows Control software as a package to replace a traditional keyboard and mouse.
Get PCEye Mini together with Gaze Point and Gaze Viewer, our entry level software offering a simple way to learn and document the use of eye tracking and gaze interaction.
The PCEye Mini, together with the zoom functionality of Windows Control, allows you to hit even the smallest targets on the screen with almost pixel precision.
Our easy, one-time calibration is helpful for those who have difficulties keeping focus on a calibration target. It’s an excellent way for beginners to get started with eye tracking.
The trackbox is the imaginary volume within which you can move your head without losing accuracy or interrupting the eye tracking session. PCEye Mini’s large trackbox size gives you the freedom to sit or lie down comfortably while maintaining eye tracking performance.
The PCEye Mini provides higher quality tracking results compared to other eye trackers, regardless of most lighting conditions, eye color, or eyewear (glasses or contact lenses).
The PCEye Mini attaches perfectly to most laptop and desktop monitors, with an excellent performance on screens up to 19’’ in size. Processing takes place within the PCEye Mini eye tracker itself, enabling you to use both old and low performance PCs.
|Technical specifications||PCEye Mini|
|Medical Device||93/42/EEC (MDD) - Medical Device Directive, as amended by 2007/47/EC|
Eye tracking (gaze point, eye position, etc.) IR image stream at 30 Hz
|Power Consumption||<1.5W typical average|
|Eye Tracker Processing Unit||IS-4B Tobii EyeChip™ with fully embedded processing|
|Track Box2 Dimensions at 65 cm (26’’) Distance (w x h)||>35 cm × 30 cm ellipse
>14” × 12”
|Working Distance||45 cm – 85 cm
18” – 33″
|Screen Size3||Recommended up to 19”|
|Detected Gaze Interaction >30 Hz||98% for 95% of population4|
(Former Tracking Robustness)
<0.2° for 65% of population
<0.25° for 80% of population
<0.3° for 90% of population
<0.4° for 95% of population
|Max Head Movement Speed (Eye Position)||40 cm/s (15.7 in/s)|
|Max Head Tilt||>20°|
|Max Yaw, Pitch||25°|
|Visibility||Low visibility (850 nm)|
|Flicker||No flicker (30 Hz constant)|
|Gaze Latency / Gaze Recovery||25 / 100 ms|
|Unit Size (Length × Height × Depth)||170 mm × 18 mm × 13 mm
6.69“ × 0.71“ × 0.51“
|Weight||59 g (2.1 oz)|
|Compatible Operating Systems||64-bit Windows 7, 8.1 or 10|
|System Requirements (Recommended Minimum)||Processor: Dual-Core 2.0 GHz
Memory: 4 GB RAM
Hard Disk: 450 MB available
USB Interface: USB 2.0 or higher
|Compatible Tobii Dynavox Software6||Windows Control (or Classic Tobii Gaze Interaction Software), Gaze Point, Gaze Viewer, Virtual Remote, Communicator 5, Snap Core First, Snap Scene, ALL, Boardmaker Student Center|
1 USB2 requirements: USB 2.0 BC1.2 provides Tobii IS4 with sufficient power and signal bandwidth.
2 Trackbox describes the space in front of the eye tracker where the user must have at least one eye to get gaze data. The eye tracker is at a 20° angle looking upward from below the screen.
3 For larger than recommended screens, the limiting parameter is the quality of the gaze accuracy at the upper corners of the screen when the user sits close to the screen.
4 The degree-of-accuracy and precision over percentage-of-population numbers result from extensive testing across representatives of the whole population. We have made use of hundreds of thousands of diagnostic images and performed tests on approximately 800 individuals with different conditions, vision, ethnicities, everyday dust, smudges or blemishes around their eyes, with eyes out of focus, etc. This has resulted in a much more robust and high performance eye tracking experience and a much more realistic representation of true performance across the whole population, not just in a mathematically "ideal" scenario.
5 The "ideal" degree numbers are the previous standard of measuring accuracy and precision, both previously from Tobii and presently from all eye tracking competitors. Though "ideal" numbers are useful to get a general feel about comparative quality and performance, they are not applicable to real world usage in the same way as quantitative degree-of-accuracy and precision over percentage-of-population numbers based on extensive testing across representatives of the whole population.
6 All software titles are not compatible with each other, see the respective software product page for more information.
WARNING: The PCEye Mini should NEVER be mounted to glass. For this reason, it is not recommended for use with a 2 in 1 laptop/tablet.